In our experience monitoring the VOC's in a laboratory is one of the most underrated QC test that people conduct.
IVFSynergy find high levels of VOC's when we test facilities during servicing or fault finding in a range of equipment that initially really surprise our clients.
High concentrations of these are widely regarded as damaging for embryo development.
Portable Handheld VOC Monitor with three second response time.
DS-1018-05 02/16 Sensor
Gas Monitor Range Resolution Response Time T90 VOCs 0 to 999.9 ppm 1,000 to 15,000 ppm 0.1 ppm 1 ppm < 3 s < 3 s
Weight 26 oz (738 g)
Photoionization sensor with standard 10.6 eV or optional 9.8 eV or 11.7 eV lamp Rechargeable, external field-replaceable Lithium-Ion battery pack
Running time 16 hours of operation (12 hours with alkaline battery adapter)
VOCs as ppm by volume (mg/m3)
Date, time, temperature Alarms 95dB at 12" (30 cm) buzzer and flashing red LED to indicate exceeded preset limits
Datalogging Standard 6 months at one-minute intervals Calibration Two-point or three-point calibration for zero and span.
Calibration memory for 8 calibration gases, alarm limits, span values and calibration dates Sampling Pump
Internal, integrated flow rate at 500 cc/mn
Sample from 100' (30m) horizontally or vertically Low Flow Alarm Auto pump shutoff at low-flow condition Communication & Data Download
Download data and upload instrument set-up from PC through charging cradle or optional Bluetooth™
Wireless data transmission through built-in RF modem Wireless Network Mesh